Turbine flowmeter maintenance should note the following:
1, when used, should keep the measured medium clean, free of fibers and particles and other impurities.
2, the turbine flow sensor in the beginning of use, the sensor should be slowly filled with media, and then open the outlet valve (valve should be installed in the back of the flow meter), the sensor is no longer in the state of the medium by the impact of high-speed fluid.
3, turbine flow sensor maintenance cycle is generally six months. When handling cleaning, be careful not to damage the parts of the measuring chamber, especially the impeller. When assembling, please look at the position of the guide and the impeller.
4, the turbine flow sensor when not in use, should clean the internal media, dry and after the sensor at both ends with a protective cover to prevent dust into, and then placed in a dry place to save.
5, with the filter should be regularly cleaned, when not in use should be cleaned of the internal media, with the same sensor, plus dust cover, placed in a dry place to save.
6, before the sensor is installed, use the mouth to blow or hand the impeller, so that it quickly shows whether the display, when there is a display when the sensor. If no display, should check the relevant parts, troubleshooting.