Ultrasonic level gauge installation see below:
⑴ ultrasonic level gauge should be placed on the top of the container.
⑵ choose an appropriate method to install the ultrasonic level gauge.
⑶ ultrasonic transmitter and receiver should be parallel with the liquid level.
⑷ ultrasonic transceiver sensing surface should be avoided near the container side wall.
⑸ To avoid vibration or shaking leading to loose gauge, please be careful to install.
⑹ the same container do not install multiple ultrasonic level gauge, ultrasonic interference will lead to measurement errors.
⑺ installation should avoid direct sunlight, outdoor installation should be protective cover.
⑻ installation location should avoid strong vibration area, such as installed in the light vibration area, should be installed rubber shock absorber.
⑼ installation location should be away from the feed port and can not encounter obstacles.
(10) If the flange is installed, please pay attention to the relationship between the height of the riser and the inner diameter of the riser.
⑾ installation interface requirements of the opening size is large enough, when the flange installation, the flange under the length of the nozzle to be designed for 10 meters, 12 meters range of the meter, take over the length should not be greater than 15CM, select DN80 above the law Orchid.
For 15 meters, 20 meters, 30 meters and 40 meters range of the meter, take over the length should not be greater than 20CM, select DN200 above the flange.
For the 8 meter range of the meter, the length of the nozzle no requirement, may be designed to eliminate the impact of blind spots, and select the DN65 above the flange.
For the level of 8 meters below the scale, no length on the length of the pipe can be properly designed to eliminate the impact of blind spots, and select the DN65 above the flange.