Germany e + h mass flow meter use
Germany e + h mass flow meter for the measurement of liquid and gas mass flow, but also can measure the fluid density and temperature. The measured values can be used to calculate other variables such as volume flow, and the various characteristics of the various fluids can be measured, for example:  · Chocolate, condensed milk, liquid sugar · oil, fat · acid, Paints, solvents and detergents • Drugs, catalysts, inhibitors • Suspensions, gases, liquefied gases, etc.
Germany e + h mass flow meter installation
Please note the following: · Installation does not require special fixed bracket, the instrument container type structure can overcome the influence of external force, such as the second chamber structure · measuring tube vibration frequency is high, to ensure that the measurement is not affected by the vibration of the pipeline. As long as there is no cavitation phenomenon, the measurement is not affected by valves, bending, tee and other fittings. · To protect the pipe, it is recommended to take the appropriate support when the sensor is too heavy. When measuring the entrainment in the tube, the measurement error is generated and the following mounting position should be avoided: • The flowmeter is installed at the highest point of the pipe and will cause the air to accumulate. • The flowmeter is installed directly at the downward vent pipe outlet. Cavitation phenomenon, because the cavitation phenomenon will affect the vibration of the measuring tube. Under normal conditions to measure the water characteristics similar to the fluid does not require special measures. When the liquid boiling point is low (hydrocarbon, solvent, liquefied gas) or on the suction pipe, the system should have a high enough pressure. To ensure that the pressure is not lower than the vapor pressure, that is, the liquid will not boil. To ensure that the liquid does not produce gasification. Therefore, the best mounting position of the sensor is: • Downstream of the pump (to avoid vacuum) • Bottom of the vertical pipe Installation direction Vertical installation It is advisable to allow the fluid to flow upwards to sink the entrained solid particles and the gas is raised by the measuring tube The The fluid in the measuring tube can be completely drained to avoid solid deposition. Horizontal installation The correct installation is the transmitter's housing vertical up or down, so that the measurement tube (single tube system) without gas and solid impurities accumulation. Do not install the sensor on the pipe, use a bracket to secure it. So as to avoid excessive tension at the process connection. The sensor mounting plate is used for mounting on a flat, wall or column.